MiniSpin® is a registered trademark of Eppendorf AG, Germany. .. >Read this operating manual completely before using the device for the first time. Also. Get Eppendorf MiniSpin plus Manual. Get all Eppendorf manuals!. Get Eppendorf MiniSpin plus Owner manual. Get all Eppendorf manuals!.
|Published (Last):||4 April 2012|
|PDF File Size:||11.95 Mb|
|ePub File Size:||3.48 Mb|
|Price:||Free* [*Free Regsitration Required]|
Extran neutral, RBS neutral. Solo allora alzare la MiniSpin ed attraverso l’apertura nella piastra del fondo, spostare il disco nella direzione della freccia, servendosi di una penna a sfera. Rotor is still spinning. Product Information Powerful and user-friendly MiniSpin and MiniSpin plus are small enough so that each workstation can be equipped with a “personal” centrifuge.
Don’t see a manual you are looking for? Stop centrifuge run and load centrifuge manuao.
Tell us about it. Tell us what’s missing. If the centrifuge makes unusual noises when started, mminispin rotor or rotor lid is not fastened correctly. No main power connection. Inspect the rotor lid regularly.
Your manual failed to upload Or point us to the URL where the manual is located. MiniSpin plus MiniSpin 14, rcf 12, rcf Ninispin.
No main power connection. Switch the device off immediately by pressing “STOP”. Please update to a current version, e.
Wait for the rotor to come to a standstill. Plug in mains cable on both sides. Rotor Assurance Plan – Visual rotor checkacc.
Centrifuge shakes during acceleration. If jinispin takes place during the eppednorf, the display switches back after 20 seconds. Installation guide — Centrifuges. MiniSpin plus MiniSpin 14, rpm 13, rpm Max. Be sure to close the tube lids tightly prior to centrifugation. Tell us about it. Load the rotor symmetrically. Rotor not loaded symmetrically. Brochure — Endless Possibilities. Inserire il cavo rete su entrambi i lati. Controllare eppenrorf fusibile del laboratorio.
Powerful and user-friendly MiniSpin and MiniSpin plus are small enough so that each workstation can be equipped with a “personal” centrifuge. If the centrifuge makes unusual noises when started, the rotor or rotor lid is not fastened correctly.
MiniSpin plus MiniSpin 14, rcf 12, rcf Max. Checking for damage Checking lid latch Checking speed sensor Verification of operating parameters acc.
Extran neutral, RBS neutral. If this takes place during the run, the display switches back after 20 seconds.
Eppendorf MiniSpin plus User Manual | 20 pages | Also for: MiniSpin
Eppehdorf test run OQ Report and certificate. Il rotore continua a girare. Non si riesce ad aprire il Manca la corrente. Wait for the rotor to come to a standstill.
With MiniSpin plus, the time can be pre-selected between eppenddorf seconds and 99 minutes, and with MiniSpin between 15 seconds and 30 minutes. Sbloccaggio di emergenza v. Our website does not support your outdated browser version. Check the mains fuse of the lab.
Attendere che il rotore si fermi. Or point us to the URL where the manual is located. Only neutral agents may be used for cleaning purposes e.
Emergency lid release see Sec. Eppendprf the rotor symmetrically. Eppendorf International Choose country. Fasten the rotor lid and close the centrifuge lid. Manca il collegamento rete.
Disconnect the centrifuge from the mains supply, remove the rotor and clean it separately. MiniSpin plus provides speed sufficient for molecular biology separations. La centrifuga vibra nell’avviamento. Cleaning of housing Checking for damage Checking lid latch Checking of rotor bowl, internal parts Checking of housing, display, keypad, stability Checking of rotor detection Checking of imbalance detection Checking of speed sensor Evaluation of stored data, Software update, System test run Discount on parts, labor, travel.